The Device Settings screen allows you to set a variety of options which modify how your ShotTrack device records data.
There are three options:
Histogram - in this mode, the device will just passively monitor the environment, recording the Peak Vector Sum of the prior 5 minutes over the length of the session.
Peak Vector Sum - in this mode, the device also records any events which exceed the thresholds set, but will only record the Peak Vector Sum at every recording interval
Waveform - in this mode, the device also records any events which exceed the threshold set, and will record the X, Y, and Z values at every recording interval.
A key consideration in choosing the last two options is that because less data is being recorded in the Peak Vector Sum mode, the record time is longer.
To see the difference you can change the record mode and see how it impacts the limits of the record time further down the Device Settings screen.
The threshold trigger for the X axis.
The threshold trigger for the Y axis.
The threshold trigger for the Z axis.
There are two options:
1000 – One thousand samples a second
2000 – Two thousand samples a second
The length of time the event recording should go for once an event has been triggered.
The G setting for your unit. Options are hardware-dependent:
2, 4, 8
10, 20, 40
How far before the event trigger that data should be captured, as a percentage of the total record time.
As an example, if you set the unit to record 32 seconds, and set the pre-trigger at 10%, then you would get 3.2 seconds of sensor data from before the event recording triggered.